To evaluate and compare the biomechanic rigidity and strength of 3 fixed-angle plates used to treat extra-articular distal radius fractures that are dorsally unstable. Volar fixed-angle plates were compared with a dorsal fixed-angle nail plate.
Three plate constructs were tested: the dorsal nail plate (DNP), distal volar radius (DVR) plate, and locking compression plate (LCP) volar distal radius plate. With anatomic, third-generation, artificial composite radii, dorsally unstable extra-articular distal radius fracture models were made by cutting a wedge osteotomy with an 8-mm dorsal gap 1 cm from the articular surface. These models were then fixed with the 3 implants by the method recommended by the manufacturer. The proximal radii of each specimen were attached to the base of a materials testing machine with a probe centered at the radial side of the lunate fossa. The specimens were loaded at a constant rate to failure under axial compression. Load and displacement were plotted graphically, and the resulting rigidities and strengths of each plate were assessed statistically.
The DVR group had significantly greater stiffness than the LCP group. The DVR group had significantly higher maximum loads than both the DNP and LCP groups. There were no significant differences in yield loads. Both the DNP and DVR groups had significantly less displacement at yield than the LCP group.
These 3 groups had similar yield loads. However, the LCP was less stiff than the DVR and had more displacement at yield than both the DVR and DNP. The yield load of all 3 implants was much higher than previously described loads for active wrist and finger motion.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Journal of Hand Surgery
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Fractures of the distal aspect of the radius: changes in treatment over the past two decades.Instr Course Lect. 2003; 52: 185-195
- Palmar plating for dorsally displaced fractures of the distal radius.Clin Orthop. 2002; 397: 403-408
- Volar fixed-angle plate fixation for unstable distal radius fractures in the elderly patient.J Hand Surg. 2004; 29A: 96-102
- The biological effect of continuous passive motion on the healing of full-thickness defects in articular cartilage.J Bone Joint Surg. 1980; 62A: 1232-1251
- Comparison of different distal radius dorsal and volar fracture fixation plates: a biomechanical study.J Hand Surg. 2003; 28A: 94-104
- Biomechanics in uniaxial compression of three distal radius volar plates.J Hand Surg. 2004; 29A: 446-451
- Palmar plate fixation of AO type C2 fracture of distal radius using a locking compression plate—a biomechanical study in a cadaveric model.J Hand Surg. 2003; 28B: 263-266
- A biomechanical comparison of a dorsal 3.5-mm T-Plate and a volar fixed-angle plate in a model of dorsally unstable distal radius fractures.J Orthop Trauma. 2005; 19: 187-191
- Surgical treatment of fractures of the distal radius with plates: a comparison of palmar and dorsal plate position.Arch Orthop Trauma Surg. 2003; 123: 333-339
- Locking versus nonlocking T-plates for dorsal and volar fixation of dorsally comminuted distal radius fractures: a biomechanical study.J Hand Surg. 2005; 30A: 756-763
- Prospective multicenter trial of a plate for dorsal fixation of distal radius fractures.J Hand Surg. 1997; 22A: 777-784
- Functional outcome and complications following two types of dorsal plating for unstable fractures of the distal part of the radius.J Bone Joint Surg. 2003; 85A: 1956-1960
- Volar versus dorsal plating in the management of intra-articular distal radius fractures.J Hand Surg. 2006; 31A: 9-16
- Fixed angle fixation of distal radius fractures through a minimally invasive approach.Tech Hand Upper Extrem Surg. 2005; 9: 142-148
- Distal radius fracture fixation with an intramedullary nail.Tech Hand Upper Extrem Surg. 2005; 9: 195-201
- Volar fixation of dorsally displaced distal radius fractures using the 2.4-mm locking compression plates.J Hand Surg. 2005; 30A: 743-749
- Structural properties of a new design of composite replicate femurs and tibias.J Biomech. 2001; 34: 773-781
- Functional outcome of unstable distal radius fractures: ORIF with a volar fixed angled tine plate versus external fixation.J Hand Surg. 2005; 30A: 289-299
- Assessment of instability factors in adult distal radius fractures.J Hand Surg. 2004; 29A: 1128-1138
- Stability assessment of distal radius fractures.Injury. 1989; 20: 208-210
- A meta-analysis of outcomes of external fixation versus plate osteosynthesis for unstable distal radius fractures.J Hand Surg. 2005; 30A: 1185-1199
- Volar fixation for dorsally angulated extra-articular fractures of the distal radius: a biomechanical study.J Hand Surg. 2006; 31A: 771-779
- Forces in the normal and abnormal hand.J Orthop Res. 1985; 3: 202-211
- Biomechanical evaluation of fixation of intra-articular fractures of the distal part of the radius in cadavera: Kirschner wires compared with calcium-phosphate bone cement.J Bone Joint Surg. 1999; 81A: 391-399
- Load transfer characteristics of the wrist. Part 1. The normal joint.J Hand Surg. 1987; 12A: 971-978
- Forearm force transmission after surgical treatment of distal radioulnar joint disorders.J Hand Surg. 1987; 12A: 196-202
- Flexor tendon forces: in vivo measurements.J Hand Surg. 1992; 17A: 291-298
Accepted: April 25, 2007
Received: October 31, 2006
Hand Innovations LLC donated internal fixation devices.
© 2007 American Society for Surgery of the Hand. Published by Elsevier Inc. All rights reserved.